Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||18 March 2011|
|PDF File Size:||14.60 Mb|
|ePub File Size:||6.55 Mb|
|Price:||Free* [*Free Regsitration Required]|
To charge the battery When the electrocardiograph is not used, connect the electrocardiograph to a wall AC socket to charge the battery.
Change the search condition and search again. The AC power lamp and battery charge lamp light.
Lit AC power lamp: Using unspecified recording paper or electrodes may cause incorrect ECG recording and accelerate deterioration of the electrocardiograph. Cleaning the Surface of the Electrocardiograph Clean the outside surface of the electrocardiograph with a soft cloth moistened with water, neutral soap or alcohol Wash with clean water and see a physician immediately. After recording, you can save the 12 lead ECG including extended sequence recording analysis result and patient information as an ECG data file in the media.
The screen is not bright and the waveform Adjust the screen brightness. Built-in lithium battery 6 years This battery backs up system settings and system clock.
Paper skews as it feeds. When performing MRI test, remove all electrodes and transducers from the patient which are connected to this instrument. The ECG analysis result is incorrect if artifact is superimposed on the notselected leads or interval.
The Main Setup screen is displayed. When the power is turned on, the Resting ECG recording screen appears.
decreto legislativo pdf – PDF Files
Retry formatting the media or check the —— media. This displays the model name, software version and analysis version of the electrocardiograph. A new fully charged battery provides at least 90 minutes continuous recording. Date and pn, Date display format, Time display format 2.
If this happens, turn the power back on to continue using the electrocardiograph. QTc interval is a corrected value based on a heart rate of 60 bpm. When using the output signal from the electrocardiograph for the synchronization signal on other instrument, always consider this time delay.
CAUTION Do not use or leave the battery pack in direct sunlight or high temperatures such as in legsilativo car in hot weather or in front of a stove. Avoid direct sprinkling, spray or moist air from a nebulizer or humidifier. Off Battery charging lamp: Select the desired item with the or function key.
Turn the power off and on again to return to the 11150 ECG recording screen. The Setup screen for that item is displayed. You can also perform automatic and manual recording during periodic ECG recording. The SD memory card is invalid.
Contact your Nihon Kohden representative. The automatic and rhythm recording result can be printed. Prints copies of the recording results. If the screen is too bright, 4.
In that period NKC or its authorized agents will repair the instrument. Select the default age group when the power is turned 3 – 5 on. Always perform equipotential grounding when required.
Remove the M3 binding head screw from the battery cover and pull off the battery cover. Select the age group by pressing the AGE key.
decreto legislativo 1150 pdf
CAL mark The height corresponds to 1 mV waveform amplitude. NOTE A note provides specific information, in the form of recommendations, prerequirements, alternative methods or supplemental information. Pull the card out straight from the SD card slot. Press decrdto function key. Back up the necessary data before formatting.
Manual Cardiofax – PDF Free Download
legislaativo This may break the cable. Adhesives, paste, felt-tip pens, and semi-dry diazo paper. CAUTION A caution alerts the user to possible injury or problems with the instrument associated with its use or misuse such as instrument malfunction, instrument failure, damage to the instrument, or damage to other property.
The Data Management – Delete screen appears. The battery charge time is at least 10 hours.
Manual Cardiofax 1150
Internal Memory Saves up to 40 files in the memory regardless of the data compression type. Store the recording paper in a cool, dry and dark place if possible. Analysis result recording cannot be 1. Lit After turning the power on, check the following items.